|Place of Origin:||FuZhou China|
|Minimum Order Quantity:||≥5PCS|
|Price:||To be Negotiable|
|Packaging Details:||Packing: Standard exporting package carton box wooden carton or As customer request|
|Delivery Time:||7-14 work days or Negotiable|
|Supply Ability:||≥100PCS Piece per month|
|Place Of Origin:||FuZhou China||Brand Name:||FZLink|
|Model Number:||ZXRB0-93||Brix Range:||0.00 To 93.00%|
|Brix Min. Div:||0.01%||Measurement Accuracy:||Brix ±0.1%|
inline process refractometer,
inline brix meter
Simplicity and best Versatile Process Management Inline concentration monitor for Brix Model ZXRB0-93
Fittings are available in a wide variety of shapes and sizes.The accuracy is ±0.1% across the entire range of 0-93% Brix.
|Measurement scale||Brix(%) (Automatic Temperature Compensation), Temperature|
Brix 0.00 to 93.0%
Temperature 5 to 100℃
|Measurement accuracy||Brix ±0.1% nD:±0.0001|
|Measurement temperature||5-100℃ (Automatic Temperature Compensation)|
Standard configuration:power line1.5m,Signal line 1.5m.Ethernet.485.Wireless standby interface
Two lines4-20mA(Brix%.Temperature ) Ethernet.485
Operation manual for Inline refractometer
1 When measuring harmful substances,please take protective measures.
2 If the instrument starts to overheat,smoke and cause abnormal smell,please turn off the power
3 Do not disassemble or modify this instrument.
4 Do not place water or sample to any other portion of the instrument except the prism surface.
5 Do not measure very hot or highly acidic samples.
6 When the power is turned on, the sample temperature should be between -10°C and 130°C.
7 When the instrument is not used for a long time,please turn off it.
Please read manual carefully and fully understand the functions and operation of each part of the instrument before using.
Before using, Check whether each part of the instrument works properly.
Make necessary inspection and adjustment according to the instruction manual.
Conditions to be followed!
Use the instrument at altitudes below 2000 meters (altitude).
Use the instrument indoors.
The using temperature shall be between 5℃ to 40℃.
Do not change the ambient temperature of the instrument suddenly.
Do not place the instrument in a place where it may be affected.
Do not use the instrument in areas with large amounts of dust and strong vibrations.
Do not put the instrument in a place where the temperature is very low.
Do not put the instrument in a damp place.
Do not put heavy objects on the instrument.
Do not install the instrument in direct sunlight or near the heating source. Do not install in the
environment of dust or exposure to corrosive gases.
When installing, please pay special attention to avoid the great impact of the instrument.
Packing: Standard exporting package carton box wooden carton or As customer request
Shipping : The products will be shipped via FEDEX DHL TNT Special-line ,sea shipping ,air transportation etc.To destination countries .
Contact Person: admin